Magnetic induction tomography (MIT) of biological tissue is used for the reconstruction of the complex conductivity distribution κ inside the object under investigation. It is based on the perturbation of an alternating magnetic field caused by the object and can be used in all applications of electrical impedance tomography (EIT) such as functional lung monitoring and assessment of tissue fluids. In contrast to EIT, MIT does not require electrodes and magnetic fields can also penetrate non-conducting barriers such as the skull. As in EIT, the reconstruction of absolute conductivity values is very difficult because of the method's sensitivity to numerical errors and noise. To overcome this problem, image reconstruction in EIT is often done differentially. Analogously, this concept has been adopted for MIT. Two different methods for differential imaging are applicable. The first one is state-differential, for example when the conductivity change between inspiration and expiration in the lung regions is being detected. The second one is frequency-differential, which is of high interest in motionless organs like the brain, where a state-differential method cannot be applied. An equation for frequency-differential MIT was derived taking into consideration the frequency dependence of the sensitivity matrix. This formula is valid if we can assume that only small conductivity changes occur. In this way, the non-linear inverse problem of MIT can be approximated by a linear one (depending only on the frequency), similar to in EIT. Keeping this limitation in mind, the conductivity changes between one or more reference frequencies and several measurement frequencies were reconstructed, yielding normalized conductivity spectra. Due to the differential character of the method, these spectra do not provide absolute conductivities but preserve the shape of the spectrum. The validity of the method was tested with artificial data generated with a spherical perturbation within a conducting cylinder as well as for real measurement data. The measurement data were obtained from a potato immersed in saline. The resulting spectra were compared with reference measurements and the preservation of the shape of the spectra was analyzed.
|Publication status||Published - 1 May 2006|
|Event||VI Conference on Biomedical Applications of Electrical Impedance Tomography - London, United Kingdom|
Duration: 22 Jun 2005 → 24 Jun 2005
|Conference||VI Conference on Biomedical Applications of Electrical Impedance Tomography|
|Period||22.06.2005 → 24.06.2005|
- Conductivity spectrum
- Magnetic induction tomography (MIT)